Static B1+ shimming
This tutorial only covers static B1+ shimming applications performed on Siemens scanners. Shimming-Toolbox currently only handles B1+ maps acquired using Siemens' standard TurboFLASH B1+ mapping sequence.
Open a Terminal and run:
Then, open the
Shimming Toolbox plugin:
Settings --> OrthoView --> Shimming Toolbox
The plugin should open as a new panel in
Download test data
From your terminal, cd to any folder and run:
This will download the example B1+ maps dataset in the folder from which you typed this command.
Convert DICOM to NIfTI
In FSLeyes, click on the
If you don't see the tab, drag the right edge of the
Shimming Toolbox panel to make all the tabs appear.
Click on Input Folder to select the downloaded
data_b1_shimming/dicomspath as an input.
Enter a subject name (e.g. "test").
(optional) Modify the output folder.
Create a Mask
Load the target anatomical image.
Select The Mask Tab.
Select Box from the dropdown.
Select the target image from the overlay, click the button Input.
Input voxel indexes for center and size. Look at the Location panel of FSLeyes to locate the center of the ROI.
(Optional) Change the output file and folder by clicking on Output File.
The output mask should load automatically in the Overlay list.
Static B1+ shimming
Navigate to the B1+ Shim Tab.
Select CV reduction in the dropdown menu (it should already be selected by default).
Select the uncombined B1+ maps from the overlay, click the button Input B1+ map.
Select the mask, click the button Input Mask.
(Optional) If you have a SarDataUser.mat VOP file (found on the scanner in C:/Medcom/MriProduct/PhysConfig.), you can locate it after clicking on input VOP file. You can then adjust the SAR factor to indicate by how much your optimized shim weights might exceed the max local SAR of a phase only shimming.
The output text file containing the shim weights should be in the Output Folder.
Manually input these shim weights on the scanner console. On Siemens scanners, input them in the
Options > Adjustments > B1 Shimwindow. Make sure to also set
System > pTx Volumes > B1 Shim modeto "Patient-specific" in the sequence parameters to ensure that the shim-weights will be applied during the acquisition.